Circuit Design Blogger
Extreme's Blog :Return to Blog
Cheap AC Current Measurement
The two cross-connected transistors act like low voltage drop diodes to generate a DC voltage which is proportional to the current in the primary of clamp adaptor (ie, the circuit under test). The recommended transistors are power germanium types such as ADZ16, AD162, AD149, ADY16, 2SD471, OC16 and OC28. This approach gives lowest voltage drop and good linearity, from 10 to 300A. Schottky power diodes can also be used but the result will not be as linear. To calibrate, wind 10 turns through the clamp adaptor’s jaws and feed a current of 20A through the winding. This is equivalent to a single turn carrying 200A. Set the trimpot to suit your multimeter, normally set to the 2V DC range. Do not calibrate for a low current otherwise accuracy at high currents will be poor.
Author: Gerard La Rooy – Copyright: Silicon Chip Electronics